There are currently no product reviews.
a solid deal - quick and without any problems.
I life in europe - with downloads no loosing time
got exactly what i ordered in a very timely manner. will use again for other manuals
I'm happy. Good quality. Very helped me with my work..............................
This is the second Manual I have ordered from owner-manuals, I give it five stars because it is exactly what I expected given the age of the equipment. So the contents look a bit aged and the pictures a bit grainy, it fulfills my needs and I am glad I can still get hold of them.
thank u so much for this manual that was so cheap that i thought it was a scam but i gambled anyway because it was too good of a deal to pass up and behold,the manual has everything and details of everything even the screws and im still amazed and very happy with my manual .so take my word and jump on it before they realize how cheap they selling thier manuals..thank you so much for taking time to read my thoughts
< CD SECTION >
PATTERN SIDE PARTS SIDE
SFR130 FE 1
CD Adjustment Method â¢ Perform the adjustments after the machine enters the test mode. â¢ Place the CD mechanism on level ground. â¢ Equipment and tools required Measuring equipment: Oscilloscope (Use the probe of 10:1) Digital Multimeter (Use it in the DC Volt range) Jitter meter (Kikusui 6235) Test Disc: TCD-782 ATD-001 1. Focus Bias Adjustment 1) Connect a digital multimeter to the test point (FE), (VC). 2) Play back the 2nd track of TCD-782. 3) Adjust SFR130 until the digital multimeter indicates 0 Â± 10 mV. 2. RF Waveform Check 1) Connect an oscilloscope to test point (RF), (VC). 2) Play back the 2nd track of TCD-782. 3) Check that the RF waveform has the maximum amplitude and the center of the wedge waveform has the clear blank. 3. Jitter Check 1) While an oscilloscope is kept connected in the same test point as in step 2. RF Waveform Check, connect the output terminal of an oscilloscope to the input terminal of the jitter meter. 2) Set the VOLT range selector of an oscilloscope to 500 mV range or lower. 3) Play back the 2nd track of TCD-782. 4) Check that the jitter meter indicates 28.0 ns or less.
DIGITAL MULTIMETER V
EYE PATTERN must be CLEAR and MAX VOLT / DIV: 500mV TIME / DIV: 0.2Âµs
OSCILLOSCOPE JITTER METER OUTPUT INPUT ns
4. Play Ability Check 1) Play back the 3rd, 8th, and 13th track of ATD-001. Check that the noise does not occur and sound skipping does not occur. 5. Laser Current Check * Do not perform this measurement unless the laser is suspected to be defective. 1) Connect a digital multimeter across the resistor R2 (10 â¦). 2) Play back the TCD-782 and check the DC voltage value on the digital multimeter. 3) Calculate the laser current (Iop) by dividing the DC voltage across R2 by the resistor value (R2 = 10 â¦). Check that the laser current (Iop) is 80 mA or less.
DIGITAL MULTIMETER V
â 43 â
Complete owner's manual in digital format. The manual will be available for download as PDF file aft…