There are currently no product reviews.
- Very good scan quality, PERFECT!
- Sehr gute scan Qualitaet, empfehlenswert!
Good site, works ok and you get what you order, no problems experienced, got my manual within a day. A++++
Original well scanned manual. Got the job done. Microwave problem found & corrected. For $5 and a new magnitron from ebay, it was a cheap and good the first shot fix. Electrical schematics allowed me to mage sure every thing else was ok before cutting and order for parts. Hard to live without.
I was very skeptical of this website, I have never downloaded manuals before. I put it on the AMEX and payed through Paypal to ensure protection. I got the manual exactly as described and now I can replace the filter capacitor for this amp. Great Price, others selling for 12.99 or more and this is the same manual. I will search out this website for other manuals. Thank you
Manual was reasonably easy to follow. I am not an engineer or know much about electronics but with the manuals help I was able to figure out the problem, identify the part required for the repair. Replacement part cost around $30. Whilst replacing the part I was telling myself, "this aint gonna work cos it seems far too easy". Took about 15 minutes to do and my plasma TV works a treat. Would never have been able to do this without the service manual.
< CD SECTION >
PATTERN SIDE PARTS SIDE
SFR130 FE 1
CD Adjustment Method â¢ Perform the adjustments after the machine enters the test mode. â¢ Place the CD mechanism on level ground. â¢ Equipment and tools required Measuring equipment: Oscilloscope (Use the probe of 10:1) Digital Multimeter (Use it in the DC Volt range) Jitter meter (Kikusui 6235) Test Disc: TCD-782 ATD-001 1. Focus Bias Adjustment 1) Connect a digital multimeter to the test point (FE), (VC). 2) Play back the 2nd track of TCD-782. 3) Adjust SFR130 until the digital multimeter indicates 0 Â± 10 mV. 2. RF Waveform Check 1) Connect an oscilloscope to test point (RF), (VC). 2) Play back the 2nd track of TCD-782. 3) Check that the RF waveform has the maximum amplitude and the center of the wedge waveform has the clear blank. 3. Jitter Check 1) While an oscilloscope is kept connected in the same test point as in step 2. RF Waveform Check, connect the output terminal of an oscilloscope to the input terminal of the jitter meter. 2) Set the VOLT range selector of an oscilloscope to 500 mV range or lower. 3) Play back the 2nd track of TCD-782. 4) Check that the jitter meter indicates 28.0 ns or less.
DIGITAL MULTIMETER V
EYE PATTERN must be CLEAR and MAX VOLT / DIV: 500mV TIME / DIV: 0.2Âµs
OSCILLOSCOPE JITTER METER OUTPUT INPUT ns
4. Play Ability Check 1) Play back the 3rd, 8th, and 13th track of ATD-001. Check that the noise does not occur and sound skipping does not occur. 5. Laser Current Check * Do not perform this measurement unless the laser is suspected to be defective. 1) Connect a digital multimeter across the resistor R2 (10 â¦). 2) Play back the TCD-782 and check the DC voltage value on the digital multimeter. 3) Calculate the laser current (Iop) by dividing the DC voltage across R2 by the resistor value (R2 = 10 â¦). Check that the laser current (Iop) is 80 mA or less.
DIGITAL MULTIMETER V
â 43 â
Complete owner's manual in digital format. The manual will be available for download as PDF file aft…